Shake bottle well before application. Ensure the refractometer prism surface is clean, wipe with distilled or RO water if necessary and dry with a cloth. Add 2 - 3 drops of solution to the prism surface and cover.
Adjust calibration mechanism to read to one of the following specifications:
1.0264 Specific Gravity
1.3394 Refractive Index
53 uS/cm at 25 deg C
Clean and dry prism surface after use. Check instrument calibration regularly, once per month is recommended. Do not add test fluid back to container. Tightly replace lid to prevent evaporation. Note: If using and ATC Refractometer, allow 45 seconds before reading.
Ingredients: Sodium chloride, RODI water